Initial measurement drift after device startup
The internal reference needs the first few ADC samples after startup to settle, which causes some measurement drift in the first +/-100 samples. See also ADS131E08S datasheet, p. 23, fig. 30.
- trigger a very short measurement after startup to let the reference settle, or
- skip the first 200 samples during analysis of a measurement taken after RocketLogger startup/reboot/reset